This item is on PRE-ORDER. Order now, Get 10% Off*
For further details, please contact us.
COVID-19 and Flu A/B Rapid Antigen Combo Test - For Self Testing (5 tests per kit)
TouchBio SARS-CoV-2 & FLU A/B Combo Rapid Antigen Test helps individuals to quickly detect SARS-CoV-2 and FLU A/B antigens in a single collected nasal swab specimen.
- Detects 3 virus antigens in a single test (SARS-CoV-2 & Influenza A and B )
- ARTG Number: 395591
- Very High Sensitivity (> 98%)
- Test Time: 15-20 mins
- Specimen Type: Nasal Swab
- Self-Test without the need for special training or equipment
- Detects SARS-CoV-2 variants including Delta & Omicron
You can download the Instruction for use (IFU) PDF here
Watch: Video Tutorial – Combo Rapid Antigen Self-Test Kits
- 5 x Test Devices
- 5 x Sterilised Swabs*
- 5 x Extraction Tubes
- 1 x Instruction for Use
- 5 x Biohazard Specimen Bags
SARS-CoV-2 & Flu A/B Antigen Combo Test Kit (Nasal)
Download the Instruction for use (IFU) PDF here
Watch "How to use"- TouchBio Combo Rapid Antigen Self-Test Kits
INTERPRETATION OF THE RESULTS
Positive test result:
If all lines are visible, Influenza A, Influenza B and SARS-CoV-2 Positive.
The shade of lines may vary, but even if a faint/weak line appears, it should be considered positive. If you have a POSITIVE result, follow the guidance. from your local State or Territory Health Department for guidance if necessary. If you are unwell seek medical assistance for SARS-CoV-2 and individuals with a positive result or who are unwell must consult a medical practitioner for follow-up clinical care for Influenza.
Negative test result:
Only one red line appears in the control region (C), and no line in the region (A), (B), and (T)
The negative result indicates that there are no Flu A/B and SARS-CoV-2 particles in the sample or the number of viral particles is below the detectable range. Even if you get a negative result, you still need to follow all public health advice on limiting the spread Covid-19 and Flu A/B.
If symptoms persist, repeat testing and consult a medical practitioner for follow-up clinical care.
Invalid test result:
No red line appears in the control region (C). The test is invalid even if there is a line on the region (A), (B), (A) and (B) or (T).
When no red line appears in the control region (C) for both tests, the test is INVALID for both Flu A/B and SARS-CoV-2.
Insufficient sample volume or incorrect procedural techniques are the most likely reasons for the control line (C) failure. Review the test procedure and repeat the test using a new test device. If invalid result continues after repeating, please contact Touch Biotechnology on the provided contact number or email for assistance.
INSTRUCTIONS FOR USE